Manual Calibration Set (CalSet)
Manual calibration unit for the
Mercury Ultratracer UT-3000
The static calibration unit is meant for
manually calibrating the
UT-3000 Mercury Ultratracer.
The calibration unit is connected to the UT-3000 Ultratracer. With a special syringe a defined volume of mercury containing gas is drawn from the calibrator and injected into the calibration port of the UT-3000. After entering the volume of the injected gas, the UT-3000 will automatically calculate the mercury mass from the volume injected and the calibration cell temperature.
The Mercury Instruments Manual Calibration Unit consists of a specially designed mercury vessel containing elemental mercury of high purity, surrounded by an aluminium jacket cooled by a thermoelectric cooler.
The temperature of the mercury vessel is exactly measured with a high precision sensor (PT 100). The electronic temperature sensor has to be connected to the UT-3000 ULTRATRACER using the cable delivered with the system.
The temperature of the calibration gas and the concentration of mercury are displayed on the UT-3000 screen before a calibration is performed.
CalSet Manual Mercury Calibration Set: Technical Specifications
|Purity of contained Hg:||99.999999%|
|Accuracy of T-Sensor:||0.01 °C|
|Power supply:||230 VAC / 50 Hz
110 VAC / 60 Hz
|Power consumption:||30 VA|
|Dimensions:||35 x 20 x 25 cm (W x H x D)|
|Weight:||approx. 4.5 kg|
|As a leading supplier of high precision analytical equipment, we strive at all times to offer top quality solutions. Our products are manufactured according to the ISO 9001 quality regulations.|